Do you have questions about this product?
Get answers from real customers and in-house experts with AnswerBox.
Q:I would like to wire this to an Oakton 1000 series conductivity controller, which has 5 terminal positions. I saw the previous answer and am a bit confused. My probe has 6 wires: clear, white, black, red, green, and brown. None are shielded.
A:The previous response you are referring to is true for the older style 35820-62 probes. The newer wiring diagram has 6 wires. The white and black are for conductivity measurement. These are connected into the conductivity cell terminals (#21 and #22 on the Cond1000). The green, red and brown are for temperature measurement and are connected to the Pt100/Pt1000 terminals (green to #18, Red to #19 and brown to #20). The clear wire is your ground/shielding wire.
Q:Could you please let me know the dimesions (diameter and heigth) of this probe?
A:The cell is 110mm long with a 12mm diameter. It will work with the 19505-20 transmitter.
Q:how to wire this to Eutech 500 transmitter. There is no guide to wire colors with thgis sensor.
A:The clear and metal-braid wires are for the conductivity. The other 3 wires (green, white and red) are for the 3-wire RTD (for temperature compensation).
Our specialists are here to help you find the best product or part available for your application. Call or Email us and we will make sure you get the right product or part for the job.