If your conductivity measurements are off, you might need a new cell, sensor, or calibration kit. Update, upgrade, or replace to get the best our of your Oakton conductivity, resistivity, or TDS meter.
Do you have questions about this product?
Get answers from real customers and in-house experts with AnswerBox.
Q:Does this probe also measure TDS? All of the meters that it is compatible with list that they measure TDS so I'm wondering if this probe can be used to measure TDS along with conductivity.
A:The probe is not what makes a meter capable to convert conductivity to TDS readings. TDS is calculated based on the conductivity measurement, this calculation is done by the meter and displayed in ppm generally. Please check the compatibility of the meter with this probe, and the capability of your meter to make the conversion to TDS.
Q:I need to measure temperature and conductivity simultaneously. Does this conductivity probe have a built in temperature sensor?
A:No, this does not have a built-in temperature sensor. That would need to be purchased as a separate probe and would depend on what meter you have.
A:Correction to the previous answer from Mike. This conductivity probe DOES include a temperature sensor.
Q:I have a Cole-Parmer Portable Waterproof Conductivity Meter 19830-51 series with a broken probe. I would like to know if the Oakton Flat-Surface Conductivity Probe works for me or if I can find the probe for my device. Mine is similar to WU-35608-00 Oakton CON 400 Waterproof Conductivity Meter.
A:Yes, you can utilize this flat surface probe or the replacement electrode that was included with the meter originally, part number 35608-50
Our specialists are here to help you find the best product or part available for your application. Call or Email us and we will make sure you get the right product or part for the job.